How to choose to blade?
Two factors are related to the result. One is the thickness of the test piece, and the other is the hardness of the table.
(1)0-60um, hard:1mm blade
(2)0-60um, soft:2mm blade
(3)61-120um, hard or soft:2mm blade
Blades (1mm11 teeth, 2mm11teeth ) -2
3m tape (19mm * 32.9m)-1
Magnifying glass-1 (Two 3V Lithium Coin Batteries are taken out. (CR 2016))
1. The cutter should penetrate the coating, cut the substrate but must not be too deep.
2. If the coating is too thick and hard to penetrate, the test is invalid. But it should be mentioned in the test report.
3. In some special situation or under special requirements, the result is relevant to the use of tap and. The tape is generally 12.7mm translucent tape, the back material is polyester film or cellulose acetate. Please take the following steps:Stick the tape over the whole grid, tear it off at the minimum angle, then get the result according to the area ratio in the tap.
All products will be checked and packaged in good condition before shipping.
After your order is confirmed, the product will be sent to you in 5-7 workdays.
If you have any questions, please feel free to contact us. We will give you a reply as soon as possible.
Multi-blade cutter spacing: 1 +0.01mm, 2+0.01mm.
Multi-blade cutter addendum straightness: ? 0.003mm ? 0.006mm.
Multi-blade cutter tooth tip width: ? 0.05mm.
Cutter spacing and cutter types: 1mm and 2mm with 11 teeth
Temperature: 23 ± 2 ° C; Relative humidity: 50 ± 5%.